Type R Thermocouple - Technical Information
Type R - Platinum-13% Rhodium vs Platinum, IEC 60584-1
Thermocouple Type R is similar to the Type S combination, this thermocouple has the advantage of slightly higher output and improved stability. In general Type R thermocouples are preferred over Type S, and applications covered are broadly identical.
Type R Thermocouples - Typical Constructions and StylesType R Mineral Insulated
Thermocouples Rugged sensors, ideal for most applications. Vast choice of terminations e.g. pot seals, cables, connectors, heads etc. Type R Thermocouples for use in Vacuum Applications Ceramic sheathed calibrated thermocouples designed specifically in vacuum furnace applications with very low leak rates Type R Miniature
Thermocouples Ideal for precision temperature measurements where minimal displacement and a fast response is required. Type R Heavy Duty
Thermocouples For industrial applications such as furnaces, kilns, ovens, boliers, flues etc. Many types of sheath material available. ATEX/IECEx Approved
Type R Thermocouples A large range of thermocouples with a variety of terminations such as pot seal, terminal heads etc.
Type R - Technical Data
|Conductor Combination||Approximate generated
EMF change in µV
per ºC change
(referenced to 0ºC) at:
|Thermocouple Output Tolerances|
to IEC 60584-1
|+ Leg||- Leg||100ºC||500ºC||1000ºC||Continuous||Short Term||Type||Class 1||Class 2||Class 3|
||8||10||13||0 to +1600ºC||-50 to +1700ºC||Temperature Range
|0°C to +1100°C
1100°C to 1600°C
±(1 +0.003 (t ⋅ 1100)°C
|0°C to +600°C
600°C to 1600°C
±0.0025 ⋅ |t|
2. Thermocouple materials are normally supplied to meet the tolerances specified in the table for temperatures above –40 deg C. These materials however, may not fall within the tolerances for low temperatures given under Class 3 for Types T, E and K thermocouples. If thermocouples are required to meet limits of Class 3, as well as those of Class 1 and/or Class 2, the purchaser should state this, as selection of materials is usually required.
3. The maximum temperature rating is dependant upon construction, size and atmospheric conditions. A more detailed guide for the maximum temperature ratings of mineral insulated thermocouples can be found here.
Cable Colour Codes for R Type Thermocouples
|International Colour Code to IEC 60584-3||ANSI MC96.1||Redundant national colour coding for thermocouple cables|
|BS 1843||DIN 43714||NFC 42324|